Anand HI-Vac (JR) Suction Units
- MS / SS Housing: Available in Mild Steel (MS) and Stainless Steel (SS).
- Electronic Sensor: To prevent splash droplets from entering the pump.
- Vacuum Gauge: 2.5 inches round vacuum gauge.
- Pump: A single head rotary vane pump.
A smaller version of our Hi-Vac suction series for use in surgical procedures where fast vacuum build-up is required.
- MAIN APPLICATIONS
- Emergency Department
- Intensive Care Unit
- OT / Surgery
- Obstetrics / Gynaecology
|Housing||MS / Stainless Steel on four castors|
|Max. Vacuum / LPM||-650 mm Hg, 30 Ltrs / min|
|Pump Type||Rotary vane pump|
|Jars||2 x 2 Ltrs. PC Jar|
|Tubing||10 mm ID x 2 mtr. (PVC)|
|Vacuum Gauge||6.25 cm dia 0-760 mm Hg|
|Power Supply||220 V AC, 50 / 60 Hz, 180 Watt|
|Noise Level||50 dB A ± 3|
|Dimensions||39 x 31 x 62 cms|
|Net Weight||25 Kg|
|Overflow Safety||Dual Mechanical|
CLEANING : After each use, lids overflow cut-off valve, suction cannulae, catheters etc. should be cleaned thoroughly. When sterilization is necessary after infective fluid has been sucked, the jars, the lids, cut-off valve and metal cannulae can be sterilized by autoclaving and the tubing, by sterilizing solution.
Direction to use
Before the unit is put on, please see that,
The plug is connected to the grounded relevant mains outlet. In case of fluctuation, use voltage
The vacuum control knob is not loose. It must be fully tightened clockwise.
The lids are tightly placed on the jars and the tube on the nozzle of the lids.
The oil container is full with lubrication oil No. 40.
CLEANING : After each use, lids overflow cut-off valve, suction cannulae, catheters etc. should be cleaned thoroughly.
When sterilization is necessary after infective fluid has been sucked, the jars, the lids, cut-off valve and metal cannulae
can be sterilized by autoclaving and the tubing, by sterilizing solution.
- 1 x Suction Unit (Rotary Pump, 35 ltrs/min)
- 2 x 2.5Ltrs Jars
- Manufacturer/Importer warranty of 1 Year against manufacturing defect
- Service and Warranty are handled by the manufacturer/importer directly for this product
- DENTAL EXPRESS will not be liable to replace the Products and the liability to replace any such Product is that of the Manufacturer/Importer
- DENTAL EXPRES and the Buyer acknowledge that the Products purchased/usedare a subject matter of original Manufacturer/Importer’s warranties only